Operating voltage: 2.3-3.3V
Battery type: 2 x AAA LRO3 batteries
Thread size: 1inch NPT
Measure accuracy: Â±1%
Maximum working pressure: 20BAR
Flow range: 15-120L / Min
Single count: 0.00-9999.9
Total cumulative count: 0-99999
Working temperature: -10Â°C ~ +60Â°C(14Â°F~140Â°F)
Standby time: 1 year
Problem solving service
*When it appears – normal flow rate but the meter does not count, when the DISPLAY button is pressed, the meter is turned on. In this case, the possible cause is: 1. the impeller is stuck or damaged; 2. the sealant material is wrapped around the impeller; Solution: Remove the meter and ensure that the impeller rotates freely.
*When it occurs – the flow rate is reduced and the meter does not count when the DISPLAY button is pressed, the meter is turned on. In this case, the reasons may be caused by: 1. The meter is partially blocked by the dried liquid; 2. Below the minimum flow rate; Corresponding solution: remove the instrument to ensure that the impeller rotates freely; 2. Increase the flow rate
* When the meter is not working properly, it may be due to a board failure or a button failure. The corresponding solution: replace the circuit board
Exceptional value. Light duty so donât bang it around. Tape the batteries in as they will pop out under shock or vibration.
Worked good for about a week then fell apart and stopped working